FAQ - FREQUENTLY ASKED QUESTIONS
TTL signal based fraction collection
FAQ: TTL signal based fraction collection
Frequently asked questions regarding the customization of the OMNICOLL fraction collector for the fraction collection based on TTL signals (optional).
Can the fraction collector accept a TTL input signal to start the collection program?
OMNICOLL could accept the TTL signal and the fraction collection is started by a 3-12 V Signal (using about 2-10mA).
If a TTL logic is needed, we could propose a customized solution of the communication module (Art. No. 6911) for this: the TTL closure 0.0-0.8 V (low) would then start the fraction collector.
LAMBDA CZ s.r.o., Lozibky 1, CZ-61400 Brno, Czech Republic - Europe; LAMBDA Instruments GmbH, Ruessenstrasse 6, CH-6340 Baar, Switzerland
Phone: +41 444 50 20 71 ; Hotline: +420 603 970 653